Citric acid based fluid neutralizer for use after an alkaline cleaning of surgical instruments, MIC instruments, rigid endoscopes, plastic implements and elastomers. Surfactant-free.
For use after an alkaline cleaning of surgical instruments.
|Our performance||Your advantage|
|Citric acid based formulation||Secure neutralization after alkaline cleaning. Prevents chalk and acid-soluble soiling|
|Efficient formulation||High efficacy with low dosing|
|Optimally matched formulation components||Broad user spectrum for acid-resistant materials from the medical area|
For cleaning and disinfection devices in the medical area, can be used with 1-3ml/l at all degrees of water hardness.
The Sekumatic™ FNZ solution must be completely rinsed off after the action; preferably use fully demineralized water (DM) for this purpose.
Aqueous citric acid solution
|Use (after commencement)||Shelf life|
Please note the following to ensure the correct storage of environmental hygiene and instrument disinfectants: The original containers (canisters, buckets or bottles) with the disinfectant concentrate (or powder) should be stored at temperatures below 25°C and kept dry and clean. A brief period below or above the given storage temperature (e.g. during transport) will not diminish the effectiveness.
The containers are to be sealed tightly after each use. Please also take care that the concentrate remaining in the container is not contaminated when preparing the application solution.
|Chemical- physical data|
|Density||20 °C||1.16 g / ml|
|pH value* (concentrate)||20 °C||ca. 2.3|
|pH-range* (solution, 1-3ml/l in DM-water)||20 °C||2.8 – 3.2|
|Conductivity (solution, 1-3ml/l in DM-water)||20 °C||2.2 – 4.6 mS / cm|
* pH-value measurements under operating conditions always by using pH-meter with temperature compensation
Neutralizer for cleaning and disinfection devices